- Load the
OMP40 Part Probe into the spindle and with a .0001" dial indicator adjust the tip of the probe to within .0002". There are
4 recessed allen screws to make this adjustment.
2” ring gauge that came with the probe on to the machine table then indicate the center or ring gauge so
the spindle is in positioned at its center. Record the machine X-Y position's
- Load the
“Part Probe” into the machine keeping it at the center of the ring gauge
the Input Button then PART SETUP Softkey
the PART PROBING Softkey
the PART PROBE DEFLECTION OFFSET Softkey
RING GAUGE METHOD Softkey
“2” for ring gauge diameter
position of the recorded XY Positions in the the X&Y Location field. You can use the STORE
MACHINE POSITION Softkey if you have not moved the machine
- Press USE
GAUGE TO GET X-Y OFFSET Softkey Then the Cycle Start button
will move and record the deflection for every 10 degrees of the Ring gauge.
to Part Probing and Probe a Circle to re-measure the ring gauge to verify the size and location of the ring Gauge.